||_ Accuracy of within +/- 0.2%
_ High Speed Sampling : 100μ sec/point
_ Maximum of 1000 point
PIV-2000 IV-Tester measures the IV curve of the sample module
of solar panels, by the trigger signal from the solar simulator.
Additionally, it also can carry out the continuous measurement
of the short circuit current (lsc).
Control signal from the IV testor triggers to emmit the light
from the solar simulator on the solar module (sample module),
and import the output from the solar module by using the bi-polar
power supply for the measurement of the maximum output (Pmax).
IV characteristics can be measured by synchronizing the current/voltage
of the sample and reference module, while varying the voltage
of the sample by the bi-polar power supply.
||FS: 2mA, 5mA, 10mA, 20mA, 50mA, 100mA,
200mA, 500mA, 1A, 2A, 5A, 10A, 20A
||FS: 0, 2V, 0, 5V, 1V, 2V, 5V, 10V, 20V,
50V, 100V, 200V, 500V
||FS: 10mA, 20mA, 50mA, 100mA, 500mA
||+/- 2.0% (*1)
||Temperature: 23 degC,
+/-3degC / Humidity: <50%
|Number of Averaging
(Proportional to the Measurement period)
|32 times (Measurement
period of 100μs)
64 times (Measurement period of 150μs)
||100μs/point (Aimed value:
40μ/point) / Short pulse type
250μs/point to 10ms/point / Long pulse type
||30 to 1000 points